Medical liquid oxygen is stored in an insulated stainless steel container that works in a similar way to a thermos flask.
These storage containers are referred to as base units and they hold the medical liquid oxygen at an extremely low temperature.
Medical liquid oxygen base units are low pressure storage containers designed to safely store more oxygen than gas cylinders can.
The temperature of liquid oxygen is -183°C and can cause painful ‘cold’ burns if it should ever come into contact with your skin.
A medical liquid oxygen system that is installed in your home will generally consist of the base unit mounted on a roller base with a separate portable unit.
IMPORTANT! Safety information for medical liquid oxygen.
In the unlikely event of a leak from the base unit, wheel the base unit outside. If possible, open the doors and windows for ventilation and allow the oxygen to disperse.
In the event of accidental contact with the skin:
Refer to the following guides
Your child’s oxygen cannula or mask are important pieces of equipment.
A cannula can easily become blocked by nasal secretions and mucus. Both cannula and masks should therefore be kept as clean as possible.
You should also check the oxygen tubing daily for any obvious signs of damage such as kinking, flattening or splitting. If you notice any damage you must replace the tubing immediately.
It is important that you keep a spare cannula or mask to hand at all times.
CAUTION: Take care that tubing does not become trapped in items such as furniture, doors, babies’ cots, infants’ high chairs etc; as this will restrict or even stop the flow of medical oxygen.
|Call the 24/7
Customer Helpline Freephone number
for your region
|London:||0808 143 9991|
|North West:||0808 143 9992|
|East Midlands:||0808 143 9993|
|South West:||0808 143 9999|
Normal working hours are
Monday to Friday 8:30am - 5:00pm
All calls are recorded for
training and quality purposes
Send an email to email@example.com